Set the Analog output maximums and Analog output minimums fields matching the desired range of the analog outputs. Specify 1 to indicate channel 1 alone. Next set the analog input maximum and minimum. It’s like having the store come to you! Only one timebase block may be present in a diagram. As mentioned earlier in this section, the counters cannot be used as both encoder inputs and PWM outputs. Channels , , and are referenced-single ended.
|Date Added:||2 October 2017|
|File Size:||15.5 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Daa following are the available options 6026e valid values for this board. For example, enter 2: For example, enter [0 1 2] in the Analog input channels field and [0 0 2] in the Analog input configurations field, to set channel 0 to referenced single-ended, channel 1 to referenced single-ended, and channel 2 to differential.
Furthermore, a value of 0 for frequency is also not acceptable. We appreciate your interest in using our InstraView website! The second differential input’s positive line must be connected to channel 1, and negative line to channel 9. Dq to this page. Specify [0, 4, 5] to indicate that lines 0, 4 and 5 are to be configured as digital outputs.
The driver would simply saturate the value at the limits. The supported analog input configurations are Referenced singled-ended 0Non-referenced singled-ended 1Differential 2.
National Instruments DAQCardE – In Stock, We Buy Sell Repair, Price Quote
For example, enter 0: The tenth differential input’s positive line must be connected to channel 17, and negative line to channel All logos and trademarks are properties of their respective companies. For devices with 8 channels: Note that there are no spaces in the line. In addition, an immediate block for the same type of channel can not be used even if they use different resources.
Set the Encoder input channels field to all the encoder channels that will be used on the board for the current diagram.
Ni Daqcarde PCMCIA Multifunction DAQ National Instruments *tested* | eBay
See the description of the HIL Initialize 606e2 for details on these modes. Attempting to do so would produce an error message. Mercury Drive Champaign, IL Valid Email Required Item to Inspect: In order to change the analog input configuration for the selected channels in the Analog Inputs tab of the HIL Initializeset the Analog input channels field to the analog input channels on the board for the current diagram, and then set the Analog input configurations field to the desired configuration.
In order to have analog outputs set to a particular voltage when the model is loaded or unloaded, the analog outputs must be daa on the HIL Initialize block’s Analog Outputs tab.
As mentioned earlier in this section, the counters cannot be used as both encoder inputs and PWM outputs. Specify  to indicate channel 0.
Ni Daqcard-6062e PCMCIA Multifunction DAQ National Instruments *tested*
Multiple options are entered on the same line separated by semicolons. Buying used equipment doesn’t always have to be a shot in the dark.
However, attempting to command a value which is out of range would not produce an error. All of the options follow the form:. Specify 1 to indicate channel 1 alone. Please feel free saq contact us at sales artisantg. Stock Quick Quote. Also, the Set PWM output parameters at model start checkbox should always be checked.
Differential for the first four channels, referenced single-ended for the next four channels. Hence, a scalar value will apply to all channels specified in the Digital output channels field.